How to check leakage for CHN?
1. please check the gas leakage of furnace and tank , after then check the leakage of whole gas circuit:
In software click “function-diagnosis” and then click “tank “to check tank gas leakage. Next click “furnace” to check furnace leakage . At last click “whole” to check whole gas circuit
2.Observe the flow of oxygen with the flowmeter
Click” function-system debugging-SV6-go” here the oxygen flow is 4.0L/min .If not , please find AS8 to adjust it to 4.0L/min
Click” function-system debugging-SV5-go” here the oxygen flow is 1.5L/min .If not , please find AS7 to adjust it to1.5L/min
Click” function-system debugging-SV4-go” here the oxygen flow is 0.5L/min .If not , please find AS6 to adjust it to 0.5L/min
3.When you start analysis , please observe the value of pressure 1 , during burning 3 , it must to reach 155000Pa ,
4.Please observe the piston’s position , open the left board with a 2 cm gap , then observe whether the piston can reach the top of the tank or not.
How to recalibrate the heat capacity of calorimeter?
Please re-calibrate the Heat Capacity as follows:
1. Select the mode to Heat Capacity determination
Select “Heat Capacity” in [setting] - [system setting] - [analysis content], input “Bomb No.” and Calorific Value of “Benzoic Acid”, then click “OK” to save it.
2. Do 3 trail tests of Benzoic acid
(1) Do 5 tests of Benzoic acid to get calorific value
If the value (the maximum value subtracts the minimum value) ≤40J, you can calculate the Heat Capacity: Enter [functions] - [Calculate Average Heat Capacity], select the 5 tests’ results, select “recalculate” and “save heat capacity”.
(2) If the value (the maximum value subtracts the minimum value)>40J, please do one more test.
a. In the 6 tests’ results, select 5 tests’ results which can reach requirement of value (the maximum value subtracts the minimum value≤40J). Then use the 5 results to calculate Heat Capacity: Enter [functions] - [Calculate Average Heat Capacity], select the 5 tests’ results, select “recalculate” and “save heat capacity”.
b. In the 6 tests’ results, if any 5 results cannot reach requirement of value (the maximum value subtracts the minimum value) ≤40J, please redo 5 tests again.
(3) Enter [system setting] - [setting] first and select the mode to calorific value determination. Then you can test the standard sample and compare the result with standard value.
How to set the program of MF6000?
For the common program, you need to set up the temperature what you want first, the default temperature is 920℃. You can set as the following steps:
1. Press "Reset" two times, then press "setup”, after input 622, press “Start”, the "Temperature Screen" will show the first temperature. After Input the first aim temperature, press "Setup", then the "Time Screen" will be modified, input the time for the first aim temperature, then press "Setup". The first temperature and time is finished.
2. Setting the second temperature and time according the same step, After Input the aim temperature, press "Setup", then input the time for the second aim temperature/"Setup" /input the third aim temperature /"Setup" /input the time for the third aim temperature/"Setup"/then save automatic. It's the end for setting
3. If you want run this program, press the common program key and start into key, it will be ok, it is easy to change this program as the above, and there is no relationship with the calibration temperature.
Whether C5500 Automatic Calorimeter can update from bench-top into vertical type
In order to save the space, a Vertical type calorimeter from CKIC is available. CKIC give the solution to change from Bench-top type to Vertical type
Step 1: Prepare the spare parts required
Name ERP No. Specification Quantity Unit Picture
Bracket 212009048 C5500B-00F 1 Pcs
Clamp 3111300038 φ60 2 Pcs
Bellows 50431383 φ54.5*48 1.5 m
Pipe 3090201003 φ15/φ11 6 m
Step 2: Adjust the horizontal of the bracket
The horizontal of the bracket is very important to keep the stable performance of the calorimeter. Put the spirit level on the bracket, adjust the
four feet on the floor to reach the horizontal.
Step 3: Load the chiller and controller
The chiller load on the bottom while the controller on the top Controller Chiller.
Step 4: Follow the number to change the pipe from the side to the top
Each hole from Side(B) to Top(A) has already been marked with numbers 1/2/3/4. Take out the pipe from Side(B) and follow the numbers to connect to inside of Top(A).
Step 5: Cut the pipe and hidden into the bellows
Step 6: Connect the pipe from the chiller to the controller
Follow the number again to connect the pipe from Top(A) from the chiller to Side(C) from the controller.
AF4000 Ash Fusin Determinator can not get the images
For the instrument can not get the images,please follow below procedures:
1.prepare 5 pyramids ,get they ready in the tray,and click "reset",there will be 1 pyramid in view.
2.click "pyramid positioning" in the software
3.adjust the camera and the mirror,make this pyramind between two red line,and insure the above surface of the tray between two green lines,and the pyramind height should more than 2/3 of the image.
Then problem will be solved.
Inquiry for Glass of Ash Fusion?
For the spare parts, you can check the picture and ERP No. in spare part module of our website (www.ckic.net/spare-parts), We also have online service to serve you in time.
For the price, our sales manager will contact you as soon as possible.
Any questions, please feel free let us know.